- Open Access
Narrow-diameter implants with conical connection for restoring the posterior edentulous region
© The Author(s). 2016
- Received: 24 May 2016
- Accepted: 27 July 2016
- Published: 5 August 2016
The objective of this retrospective study was to show results from platform-switched narrow-diameter implants in the posterior edentulous region, which we followed up for more than 1 year after functional loading.
Ninety-eight narrow implants were inserted into 66 patients. After healing, fixed implant-supported prostheses were delivered to the patients, and Periotest and radiographic examinations were performed. After the first year of loading, the implant outcome was again evaluated clinically and radiographically using the Periotest analysis. Crestal bone loss and Periotest values (PTVs) were used to evaluate the effect of surgery, prosthesis, implant, and a host-related factor. A general linear model was used to statistically detect variables statistically associated with crestal bone loss and Periotest value.
We followed up on the implants over 1 to 4 years after loading; their survival rate was 100 %, and pronounced differences from PTVs were noted among jaw location, bone quality, and loading period. No difference was detected in bone loss among the variables studied. Bone loss after functional loading was 0.14 ± 0.39 mm. The stability value from the Periotest was −3.29 ± 0.50.
Within the limitations of this study, judicious use of platform-switched narrow implants with a conical connection must be considered an alternative for wide-diameter implants to restore a posterior edentulous region.
- Dental implant platform switching
- Conical dental implant-abutment connection
- Narrow-diameter implants
- Posterior edentulous
Edentulous alveolar ridges less than 5 mm wide require horizontal augmentation or expansion to position regular implants and produce the necessary bone quantity (at least 1 mm of bone on the buccal and oral side) . Various surgical techniques including ridge splitting, ridge expansion, lateral augmentation, and horizontal distraction osteogenesis are necessary to increase bone availability in the narrow alveolar ridge. However, these procedures can be problematic in terms of cost-effectiveness, surgery time, and healing time. Katranji et al. demonstrated that the average cancellous thickness ranges from 2.86 to 4.54 in the edentulous mandible and maxilla in the molar region, and the average cancellous thickness ranges from 2.12 to 3.11 in the edentulous maxilla and mandible in the premolar region . Approximately, 3.0- to 4.5-mm-diameter implants should be applied to the edentulous area for bicortical stabilization by engaging both buccal and lingual cortical bones for consistency with Katranji’s report. It is recommended that wide diameter and longer length of implants are installed in the posterior region given certain biomechanical aspects. However, a guarantee of long-term success for wide implants is controversial. Shin et al. reported that the survival rate for a regular-diameter implant is higher than that for wide-diameter implants . They noted that this result might be related not only to specific implant design features but also to the relative relationship of the implant to the host bone dimensions . Typically, 3.75-mm diameter implants are considered standard or regular, which is below and above implants with narrow and wide diameters, respectively . Although various reasons are mentioned for diameter selection, no study clearly supports such explanations. In choosing the implant diameter for the posterior region, the most important factors are the emergence profile, the residual bone width, and occlusal force. Occlusal force decreases with age , and likely, adequate occlusal force is proportional to the remaining bone quantity, especially in elderly patients with a narrow ridge [6, 7]. Such observations indicate a varied occlusal force that corresponds to age, race, and sex. A retrospective study using 202 narrow-diameter implants reported a 96 % success rate . An additional study that used 30 narrow-diameter single implants and followed up for 3 to 7 years reported that one fixture was fractured .
However, the literature is sparse on reports that evaluate narrow implants in the posterior area. The aim of this study was to evaluate the clinical outcome, survival rate, bone loss, and mechanical and prosthetic complications for narrow implants as well as follow up for more than 1 year after functional loading in the posterior edentulous region.
The crowns or bridges were glued onto the abutments using provisional cement (TempBond®; Kerr Co.,USA). A radiographic examination was performed using a paralleling technique with a device (Dentsply Rinn, Elgin, IL, USA) and a digital imaging software system (PiViewSTAR® (INFINITT, Korea)). Intraoral digital radiographs were generated at the prosthesis delivery and 6-month intervals after loading. The crestal bone levels were measured using the vertical distance from a fixture reference point from the bone level. The reference point was the most coronal point to the vertical portion of the fixture . The distal and mesial sides of each implant were measured, and a mean value per implant was then calculated. The geometry of the implant was used to assess the distortion of the images. Crestal bone levels that were coincident with the top of the fixture or coronal to it were given the value 0. An electronic mobility-testing device (Periotest; Siemens AG, Bensheim, Germany) was used for the measurements in accordance with the manufacturer’s recommendations on the final follow-up date after final prosthesis removal. The measurements were performed by a single rater, who was blind to the treatment condition. The electronic mobility-testing device provided reproducible data related for the bone-implant complex . The Periotest values (PTVs) ranged from −8 to +50. A range from −8 to +9 corresponded to zero on the Miller index . Statistical significance was p < 0.05. Bone loss and PTVs were classified according to the variables. A general linear model was used to discern variables associated with bone loss and PTVs. Statistical data analyses were conducted using SAS version 9.1 (SAS, Cary, NC) and SPSS version 12.0 (SPSS, Chicago, IL, USA).
Distribution of implant positions
Number of implants
Number of implants
Comparison of bone loss and PTV by variable
Number of implants
Bone loss mean (SD)
PTV mean (SD)
0.15 mm (0.376)
0.15 mm (0.423)
0.13 mm (0.375)
0.18 mm (0.439)
0.11 mm (0.351)
Over 1 year
0.08 mm (0.282)
Over 2 years
0.21 mm (0.535)
Over 3 years
0.16 mm (0.439)
Over 4 years
0.11 mm (0.259)
0.16 mm (0.418)
0.08 mm (0.245)
Reason for tooth loss
0.13 mm (0.384)
0.29 mm (0.488)
Association between variables and PTV using the regression model
Pr > |t|
Pr > |t|
Reason for tooth loss
Association between variables and bone loss using regression model
Pr > |t|
Pr > |t|
Reason for tooth loss
In this study, no implants were lost, and two implants had biomechanical problems during the follow-up period. There is a risk for restoring a posterior region using narrow-diameter implants due to the high masticatory force in molar area . Fatigue fracture of the narrow implant body from mechanical weakening has been reported . It has been suggested that a 4-mm-wide implant has a 30 % higher fatigue resistance than a 3.75-mm-wide implant . Therefore, many previous studies reported that wide implants provide better biomechanical characteristics , but under certain circumstances, it is difficult to use regular or wide implants. When wide implants are installed in narrow ridges, many clinicians must bone graft around the fenestrated implant surfaces. However, it is postulated that peri-implant grafted bone will be resorbed if the grafted bone does not have an optimal osteogenesis period.
Bone augmentation procedures are often necessary to enlarge the bone width and facilitate regular- or wide-implant positioning. Autogenous bone grafts require complex surgical techniques, and additional risks must be considered. Using narrow implants gives an unskilled clinician surgical freedom and is applicable in patients without the bone width required for regular-diameter implant installation. Although no fenestration has been reported around the implants when narrow implants are used, bone grafts can be necessary to clear a food bolus. In this study, bone grafts over the alveolar bone using slow-resorbing material such as Bio-Oss® (Geistlich, Wolhusen, Switzerland) were used to restore the optimal alveolar bone width for the buccinator mechanism .
Using narrow implants reduces the chance of bone dehiscence or fenestration during a flapless surgery. It also prevents lingual dehiscence in the mandibular second molar area during preparation. Given the decreased width of the drills and implants, osteotomy preparation implies less risk of overheating the bone.
The fixture-abutment connection type is also important for implant longevity. Quek et al. reported that narrow-diameter implants are more easily broken than wider implants because they have a narrow platform diameter . However, different results are expected from implants with an internal connection. Certain studies have shown that the biomechanical stability of internal conical connection implants is better than in butt-joint implants [22, 23]. Herein, we used implants with a conical connection, and the force on implants with a conical connection was not focused on a screw but a connection.
In the previous study, the effect of the joint design on the fatigue strength and failure mode in the conical connection system was significantly better for the butt-joint system . Therefore, it would be difficult to apply Quek’s result  to the system used herein. In an article published by Zipprich et al., 10 implant systems that used either conical or flat-to-flat connections were compared relative to their dynamic lateral load responses under simulated clinical conditions. The clearance-fit systems produced micromovements, whereas the systems with a conical fit (Astratech® and Ankylos®) generated no movement at all . There will be no micromovement during functional load, and fewer loads on the abutment screws produce few if any screw loosening problems.
One of the pitfalls in using narrow implants is the risk of fracture to the fixture or abutment. The thickness of the fixture titanium wall is important. Where the fixture titanium is too thin around the abutment, the tendency is to lose bone upon loading. Therefore, it is important to secure sufficient fixture titanium around the abutment. A previous study showed that reinforcing the neck region is necessary in reduced-diameter Straumann® tissue level implants . Quaresma et al. reported that a conical-connection implant produces lower stress on the alveolar bone and prosthesis and greater stress on the neck portion of the abutment-prosthesis complex . It has a weak point at the neck portion of the abutment, especially when it is used in the posterior region. Herein, we observed no abutment fracture cases. We presumed that narrow fixtures were generally used in the narrow width of the bone because patients with a narrow bone width may have a weaker occlusal force than patients with normal- or wide-width bones . Interestingly, the Ankylos® system design generates the same abutment for 3.5-mm-, 4.5-mm-, 5.5-mm-, and 7-mm-diameter implant fixtures. Thus, the crown margin position is only determined by the abutment, and after healing, several millimeters of play are available to define the final emergence profile; further, the same connection size between fixture and abutment can be used for all regions in the mouth. If the bone width is wide enough for installation of a wide implant, there is no reason not to use wider implants. However, it should also be noted that sufficient bone housing around the implants may be more important than implant diameter. The premise should be biomechanical stability. In Asian patients, who usually have narrow ridges and thin gingival biotype, using narrow implants enables bicortical installation in the posterior region. However, when treating patients with severe bruxism, heavy masticatory forces and oral habits, a narrow implant is not advised. We observed one case of cement loss in a bruxing patient. The loading history of the implant and the time required for the functional adaptation of the bone to implants may be more important than the implant itself. Development of better biomechanical properties in implants will facilitate narrow implant use in the posterior region. No statistically significant association between the variables and bone loss were detected herein given the combination of the factors described above, such as platform switching and conical connection. A single factor does not produce implant treatment success. Streckbein et al. also reported that low levels of bone strain are observed where a platform switch compensates for a small cone angle in the Ankylos® system . For the PTV value, the groups with a loading period over 3 years is smaller than the group with a loading period of over 1 or 2 years, which is likely related to bone remodeling completion.
Within the limitations of this study, the prognosis for narrow implants with a conical connection in the posterior region is comparable to wider implants. Thus, narrow implants can likely be used successfully. However, further research is necessary to determine the long-term success of narrow implants in the posterior edentulous region.
The authors gratefully acknowledge the Graphic Designer, Ernest Pacheco (firstname.lastname@example.org), for supplying the beautiful illustration of implant system used in this study.
IH participated in making the conception and design of case report and carried out the acquisition of data and drafting of article. JW participated in making the conception and design of case report and carried out drafting of article. SY performed the statistical analysis and interpretation of data. YH carried out the acquisition of data and made a final approval and guarantor of manuscript. BE participated in making the conception and design of case report, carried out the acquisition of data and made a final approval and guarantor of manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Albrektsson T, Zarb G, Worthington P, Eriksson AR (1986) The long-term efficacy of currently used dental implants: a review and proposed criteria of success. Int J Oral Maxillofac Implants 1:11–25PubMedGoogle Scholar
- Katranji A, Misch K, Wang HL (2007) Cortical bone thickness in dentate and edentulous human cadavers. J Periodontol 78:874–8View ArticlePubMedGoogle Scholar
- Shin SW, Bryant SR, Zarb GA (2004) A retrospective study on the treatment outcome of wide-bodied implants. Int J Prosthodont 17:52–8PubMedGoogle Scholar
- Comfort MB, Chu FC, Chai J, Wat PY, Chow TW (2005) A 5-year prospective study on small diameter screw-shaped oral implants. J Oral Rehabil 32:341–5View ArticlePubMedGoogle Scholar
- Cehreli MC, Akca K (2004) Narrow-diameter implants as terminal support for occlusal three-unit FPDs: a biomechanical analysis. Int J Periodontics Restorative Dent 24:513–9PubMedGoogle Scholar
- Jonasson G, Kiliaridis S (2004) The association between the masseter muscle, the mandibular alveolar bone mass and thickness in dentate women. Arch Oral Biol 49:1001–6View ArticlePubMedGoogle Scholar
- Galo R, Vitti M, Santos CM, Hallak JE, Regalo SC (2006) The effect of age on the function of the masticatory system—an electromyographical analysis. Gerodontology 23:177–82View ArticlePubMedGoogle Scholar
- Lazzara R, Siddiqui AA, Binon P et al (1996) Retrospective multicenter analysis of 3i endosseous dental implants placed over a five-year period. Clin Oral Implants Res 7:73–83View ArticlePubMedGoogle Scholar
- Polizzi G, Fabbro S, Furri M, Herrmann I, Squarzoni S (1999) Clinical application of narrow Branemark System implants for single-tooth restorations. Int J Oral Maxillofac Implants 14:496–503PubMedGoogle Scholar
- Astrand P, Engquist B, Anzen B et al (2004) A three-year follow-up report of a comparative study of ITI dental implants and Branemark system implants in the treatment of the partially edentulous maxilla. Clin Implant Dent Relat Res 6:130–41View ArticlePubMedGoogle Scholar
- Oh JS, Kim SG, Lim SC, Ong JL (2009) A comparative study of two noninvasive techniques to evaluate implant stability: Periotest and Osstell Mentor. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 107:513–8View ArticlePubMedGoogle Scholar
- Schulte W, Lukas D (1993) Periotest to monitor osseointegration and to check the occlusion in oral implantology. J Oral Implantol 19:23–32PubMedGoogle Scholar
- Romanos GE, Nentwig GH (2000) Single molar replacement with a progressive thread design implant system: a retrospective clinical report. Int J Oral Maxillofac Implants 15:831–6PubMedGoogle Scholar
- Ferrario VF, Sforza C, Serrao G, Dellavia C, Tartaglia GM (2004) Single tooth bite forces in healthy young adults. J Oral Rehabil 31:18–22View ArticlePubMedGoogle Scholar
- Allum SR, Tomlinson RA, Joshi R (2008) The impact of loads on standard diameter, small diameter and mini implants: a comparative laboratory study. Clin Oral Implants Res 19:553–9View ArticlePubMedGoogle Scholar
- Rangert B, Krogh PH, Langer B, Van Roekel N (1995) Bending overload and implant fracture: a retrospective clinical analysis. Int J Oral Maxillofac Implants 10:326–34PubMedGoogle Scholar
- Bahat O, Handelsman M (1996) Use of wide implants and double implants in the posterior jaw: a clinical report. Int J Oral Maxillofac Implants 11:379–86PubMedGoogle Scholar
- Berry DC (1979) The buccinator mechanism. J Dent 7:111–4View ArticlePubMedGoogle Scholar
- Tarnow DP, Cho SC, Wallace SS (2000) The effect of inter-implant distance on the height of inter-implant bone crest. J Periodontol 71:546–9View ArticlePubMedGoogle Scholar
- Rompen E, Domken O, Degidi M, Pontes AE, Piattelli A (2006) The effect of material characteristics, of surface topography and of implant components and connections on soft tissue integration: a literature review. Clin Oral Implants Res 17(Suppl 2):55–67View ArticlePubMedGoogle Scholar
- Quek CE, Tan KB, Nicholls JI (2006) Load fatigue performance of a single-tooth implant abutment system: effect of diameter. Int J Oral Maxillofac Implants 21:929–36PubMedGoogle Scholar
- Khraisat A, Stegaroiu R, Nomura S, Miyakawa O (2002) Fatigue resistance of two implant/abutment joint designs. J Prosthet Dent 88:604–10View ArticlePubMedGoogle Scholar
- Merz BR, Hunenbart S, Belser UC (2000) Mechanics of the implant-abutment connection: an 8-degree taper compared to a butt joint connection. Int J Oral Maxillofac Implants 15:519–26PubMedGoogle Scholar
- Zipprich H, Weigle P, Lauer H-C, Lange B (2007) Micro-movements at the implant-abutment interface: measurements, causes and consequenses. Implantologie 15:31–45Google Scholar
- Akca K, Cehreli MC, Iplikcioglu H (2003) Evaluation of the mechanical characteristics of the implant-abutment complex of a reduced-diameter morse-taper implant. A nonlinear finite element stress analysis. Clin Oral Implants Res 14:444–54View ArticlePubMedGoogle Scholar
- Quaresma SE, Cury PR, Sendyk WR, Sendyk C (2008) A finite element analysis of two different dental implants: stress distribution in the prosthesis, abutment, implant, and supporting bone. J Oral Implantol 34:1–6View ArticlePubMedGoogle Scholar
- Streckbein P, Streckbein RG, Wilbrand JF et al (2012) Non-linear 3D evaluation of different oral implant-abutment connections. J Dent Res 91:1184–9View ArticlePubMedGoogle Scholar