- Open Access
Recent advances in the reconstruction of cranio-maxillofacial defects using computer-aided design/computer-aided manufacturing
© The Author(s). 2018
- Received: 30 December 2017
- Accepted: 16 January 2018
- Published: 5 February 2018
With the development of computer-aided design/computer-aided manufacturing (CAD/CAM) technology, it has been possible to reconstruct the cranio-maxillofacial defect with more accurate preoperative planning, precise patient-specific implants (PSIs), and shorter operation times. The manufacturing processes include subtractive manufacturing and additive manufacturing and should be selected in consideration of the material type, available technology, post-processing, accuracy, lead time, properties, and surface quality. Materials such as titanium, polyethylene, polyetheretherketone (PEEK), hydroxyapatite (HA), poly-DL-lactic acid (PDLLA), polylactide-co-glycolide acid (PLGA), and calcium phosphate are used. Design methods for the reconstruction of cranio-maxillofacial defects include the use of a pre-operative model printed with pre-operative data, printing a cutting guide or template after virtual surgery, a model after virtual surgery printed with reconstructed data using a mirror image, and manufacturing PSIs by directly obtaining PSI data after reconstruction using a mirror image. By selecting the appropriate design method, manufacturing process, and implant material according to the case, it is possible to obtain a more accurate surgical procedure, reduced operation time, the prevention of various complications that can occur using the traditional method, and predictive results compared to the traditional method.
- Computer-aided design/computer-aided manufacturing
- Three-dimensional imaging
- Cranio-maxillofacial defect
- Reconstructive surgical procedures
- Custom implant
- Patient-specific implant
The reconstruction of complex cranio-maxillofacial defects is challenging due to the unique anatomy, the presence of a vital structure, and the variety of deficits [1, 2]. The reconstruction of congenital or acquired cranio-maxillofacial defects due to congenital abnormalities, post-trauma, tumor resection, and infection requires both functional and esthetic considerations [3, 4].
Computer-aided design (CAD) is the process of creating, modifying, analyzing, or optimizing a design using computer system. Computer-aided manufacturing (CAM) is the process of planning, managing, or controlling manufacturing using computer system .
The advantages of CAD/CAM technology include improved accuracy of esthetic results, restoration of large and geometrically complex anatomical defects, reduction of operative times, more accurate fitting of implants, overcoming the disadvantages of autogenous bone grafts, and performing resection and reconstruction in one step [10, 11]. The 3D printing technique in the cranio-maxillofacial area surgery includes contour models that are accurate replicas of patient-specific anatomy, guides that are patient-specific templates that guide precise cutting and drilling, splints defined as the replica of the virtual post-operative position of the patient structure, and implants defined as three-dimensionally printed objects that are directly implanted in the patients [12, 13].
In this paper, we will discuss the manufacturing processes using CAD/CAM, implant materials, the workflow reconstructing the cranio-maxillofacial defects, and future directions of development.
The manufacturing processes
The manufacturing processes include subtractive manufacturing, which cuts off a piece of material to form the final shape, and additive manufacturing, which builds up the material by stacking . Subtractive manufacturing, the traditional machining technique has the disadvantage in that it is difficult to make complicated shapes by computer numerical control (CNC) milling and there is a lot of material waste .
There are various kinds of manufacturing processes. The manufacturing process should be selected with consideration of the material type, available technology, post-processing, accuracy, lead time, properties, and surface quality .
The ideal material is biocompatible, easy to shape, high strength, non-toxic, inexpensive, durable, radiolucent, and lightweight [8, 22]. However, no material satisfies these conditions [22–24]. Materials include non-resorbable materials such as titanium, polyethylene, polyether ether ketone (PEEK), and hydroxyapatite (HA) and absorbable materials such as poly-DL-lactic acid (PDLLA), polylactide-co-glycolide acid (PLGA), and calcium phosphate.
Titanium is the metal of choice for manufacturing implants. It has the advantages of high strength, biocompatibility, lightweight, corrosion resistance, and the potential for osseointegration [8, 25, 26]. However, it has the disadvantage of causing scatter artifacts in CT scans .
Polyethylene includes porous polyethylene (PPE) and ultra-high molecular weight polyethylene (UHMW-PE). PPE such as Medpor (Pufex Surgical Inc., College Park, GA, USA) was used for reconstruction of the orbital floor and augmentation of the facial area . PPE is very stable and easy to shape and has tissue ingrowth through its pores [29, 30]. However, there is a possibility of infection . UHMW-PE was used for reconstruction of orbit or temporomandibular joint by making PSIs using CAD/CAM [31, 32]. Because of a solid structure, UHMW-PE can have a lower infection rate than PPE . Polyethylene has the advantage of not producing artifact because of radiolucency in CT, but it also has a disadvantage that it is difficult to control implant position after surgery [3, 32].
PEEK was used to reconstruct various craniofacial bone defects including cranioplasty [23, 33]. PEEK has similar strength and elasticity to bone and is easy to modify . It is radiolucent in CT and offers more comfort to patients, with lower thermal conductivity and lighter weight than titanium. However, it had reports of infection and foreign body reaction .
HA is used as a biocompatible scaffolding material for bone tissue engineering . It is osteoconductive and non-resorbable and shows tissue in-growth in the presence of pores, with a strong capacity to bind both hard and soft tissues . Pure HA is low in viscosity and difficult to make complex shapes, but it can be overcome with custom-made HA using CAD/CAM [37, 38].
As absorbable implants, PDLLA and PLGA are commonly used, especially in pediatric craniofacial surgery . However, foreign body reaction and the weakness of materials such as screw fracture have been reported [40, 41]. There are cases in which calcium phosphate implants have been used for the reconstruction of cranio-maxillofacial defects . These printed calcium phosphate implants have good biocompatibility and suitable biodegradation and are similar to the mineral phase of the bone, so they do not cause artifacts or interference seen in other metallic alloplasts in CT or MRI. In addition, calcium phosphate implants show less mechanical performance than titanium but are suitable as a scaffold for bone tissue growth and can be loaded with bioactive protein or antibiotics.
The modeling software used for 3D printing includes Mimics (Materialise, Leuven, Belgium), SolidWorks (Dassault Systemes, Velizy-Villacoublay, France), Amira (FEI Visualization Sciences Group, Merignac, France), Rhino (Robert McNeel & Associates, Seattle, WA, USA), and SurgiCase CMF (Materialise, Leuven, Belgium). The printing software include ZPrinter and Projet (3D Systems, Rock Hill, SC, USA) and Alaris (Objet Limited, Rehovot, Israel) .
After performing virtual surgery, including resection and reconstruction in the pre-operative image, printing of a resection guide or fabrication of the template by printing the skull model with virtual surgery [49, 50]
After printing of the skull model reconstructed symmetrically using a mirror image of the unaffected side, pre-bending of the plate, using it as a template, or fabrication of the implant directly on the skull model [51–53]
After reconstruction symmetrically using a mirror image of the unaffected side and design of the 3D implant to fit precisely to the defect, fabrication of the PSI by transferring the PSI data to CAM software [3, 8, 32, 54–57]
Digital workflows are time consuming and cannot be used for emergency procedures such as immediate post-traumatic surgery . It takes from a few days to weeks to make the PSI outside the hospital . However, with the development of 3D printers, relatively inexpensive personalized 3D printers have been introduced and the accuracy has increased, making it possible to manufacture products inside the hospital, reducing the time required. In addition, the development of professional CAD software familiar to surgeons and minimally invasive surgical procedures will provide predictable results.
By selecting the appropriate design method, manufacturing process, and implant material according to the case, it is possible to obtain a more accurate procedure, reduced surgical time, the prevention of various complications that can occur using the traditional method, and predictive results compared to the traditional method.
Availability of data and materials
The dataset supporting the conclusions of this article is included within the article.
The author wrote the manuscript and approved the final manuscript.
Ethics approval and consent to participate
Consent for publication
The author declares no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
- Rachmiel A, Shilo D, Blanc O, Emodi O (2017) Reconstruction of complex mandibular defects using integrated dental custom-made titanium implants. Br J Oral Maxillofac Surg 55:425–427View ArticlePubMedGoogle Scholar
- Kim JW, Kim DY, Ahn KM, Lee JH (2016) Surgical implications of anatomical variation in anterolateral thigh flaps for the reconstruction of oral and maxillofacial soft tissue defects: focus on perforators and pedicles. J Korean Assoc Oral Maxillofac Surg 42:265–270View ArticlePubMedPubMed CentralGoogle Scholar
- Rudman K, Hoekzema C, Rhee J (2011) Computer-assisted innovations in craniofacial surgery. Facial Plast Surg 27:358–365View ArticlePubMedGoogle Scholar
- Seok H, Kim MK, Kim SG (2016) Reconstruction of partial maxillectomy defect with a buccal fat pad flap and application of 4-hexylresorcinol: a case report. J Korean Assoc Oral Maxillofac Surg 42:370–374View ArticlePubMedPubMed CentralGoogle Scholar
- Groover MP, Zimmers EW (1984) CAD/CAM: computer-aided design and manufacturing. Prentice-Hall, Englewood Cliffs; LondonGoogle Scholar
- Smith EH (2013) Mechanical engineer’s reference book, 12th edn. Butterworth Heinemann, Oxford, England; BostonGoogle Scholar
- Toth BA, Ellis DS, Stewart WB (1988) Computer-designed prostheses for orbitocranial reconstruction. Plast Reconstr Surg 81:315–324View ArticlePubMedGoogle Scholar
- Owusu JA, Boahene K (2015) Update of patient-specific maxillofacial implant. Curr Opin Otolaryngol Head Neck Surg 23:261–264View ArticlePubMedGoogle Scholar
- Levine JP, Patel A, Saadeh PB, Hirsch DL (2012) Computer-aided design and manufacturing in craniomaxillofacial surgery: the new state of the art. J Craniofac Surg 23:288–293View ArticlePubMedGoogle Scholar
- Peel S, Bhatia S, Eggbeer D, Morris DS, Hayhurst C (2017) Evolution of design considerations in complex craniofacial reconstruction using patient-specific implants. Proc Inst Mech Eng H 231:509–524View ArticlePubMedGoogle Scholar
- Kang SH, Kim MK, Lee JY (2016) Single-tooth dento-osseous osteotomy with a computer-aided design/computer-aided manufacturing surgical guide. J Korean Assoc Oral Maxillofac Surg 42:127–130View ArticlePubMedPubMed CentralGoogle Scholar
- Jacobs CA, Lin AY (2017) A new classification of three-dimensional printing technologies: systematic review of three-dimensional printing for patient-specific craniomaxillofacial surgery. Plast Reconstr Surg 139:1211–1220View ArticlePubMedGoogle Scholar
- Lee YC, Sohn HB, Kim SK, Bae OY, Lee JH (2015) A novel method for the management of proximal segment using computer assisted simulation surgery: correct condyle head positioning and better proximal segment placement. Maxillofac Plast Reconstr Surg 37:21View ArticlePubMedPubMed CentralGoogle Scholar
- Watson J, Hatamleh M, Alwahadni A, Srinivasan D (2014) Correction of facial and mandibular asymmetry using a computer aided design/computer aided manufacturing prefabricated titanium implant. J Craniofac Surg 25:1099–1101View ArticlePubMedGoogle Scholar
- Parthasarathy J (2014) 3D modeling, custom implants and its future perspectives in craniofacial surgery. Ann Maxillofac Surg 4:9–18View ArticlePubMedPubMed CentralGoogle Scholar
- Gokuldoss PK, Kolla S, Eckert J (2017) Additive manufacturing processes: selective laser melting, electron beam melting and binder jetting-selection guidelines. Materials (Basel) 10:E672Google Scholar
- Sidambe AT (2014) Biocompatibility of advanced manufactured titanium implants—a review. Materials (Basel) 7:8168–8188View ArticlePubMed CentralGoogle Scholar
- Fina F, Goyanes A, Gaisford S, Basit AW (2017) Selective laser sintering (SLS) 3D printing of medicines. Int J Pharm 529:285–293View ArticlePubMedGoogle Scholar
- Otawa N, Sumida T, Kitagaki H et al (2015) Custom-made titanium devices as membranes for bone augmentation in implant treatment: modeling accuracy of titanium products constructed with selective laser melting. J Craniomaxillofac Surg 43:1289–1295View ArticlePubMedGoogle Scholar
- Bandyopadhyay A, Krishna BV, Xue W, Bose S (2009) Application of laser engineered net shaping (LENS) to manufacture porous and functionally graded structures for load bearing implants. J Mater Sci Mater Med 20(Suppl 1):S29–S34View ArticlePubMedGoogle Scholar
- Fafenrot S, Grimmelsmann N, Wortmann M, Ehrmann A (2017) Three-dimensional (3D) printing of polymer-metal hybrid materials by fused deposition modeling. Materials (Basel) 10:E1199Google Scholar
- Rubin JP, Yaremchuk MJ (1997) Complications and toxicities of implantable biomaterials used in facial reconstructive and aesthetic surgery: a comprehensive review of the literature. Plast Reconstr Surg 100:1336–1353View ArticlePubMedGoogle Scholar
- Zanotti B, Zingaretti N, Verlicchi A, Robiony M, Alfieri A, Parodi PC (2016) Cranioplasty: review of materials. J Craniofac Surg 27:2061–2072View ArticlePubMedGoogle Scholar
- Aydin S, Kucukyuruk B, Abuzayed B, Aydin S, Sanus GZ (2011) Cranioplasty: review of materials and techniques. J Neurosci Rural Pract 2:162–167View ArticlePubMedPubMed CentralGoogle Scholar
- Acero J, Calderon J, Salmeron JI, Verdaguer JJ, Concejo C, Somacarrera ML (1999) The behaviour of titanium as a biomaterial: microscopy study of plates and surrounding tissues in facial osteosynthesis. J Craniomaxillofac Surg 27:117–123View ArticlePubMedGoogle Scholar
- Schipper J, Ridder GJ, Spetzger U, Teszler CB, Fradis M, Maier W (2004) Individual prefabricated titanium implants and titanium mesh in skull base reconstructive surgery. A report of cases. Eur Arch Otorhinolaryngol 261:282–290View ArticlePubMedGoogle Scholar
- Shah AM, Jung H, Skirboll S (2014) Materials used in cranioplasty: a history and analysis. Neurosurg Focus 36:E19View ArticlePubMedGoogle Scholar
- Ridwan-Pramana A, Wolff J, Raziei A, Ashton-James CE, Forouzanfar T (2015) Porous polyethylene implants in facial reconstruction: outcome and complications. J Craniomaxillofac Surg 43:1330–1334View ArticlePubMedGoogle Scholar
- Wellisz T (1993) Clinical experience with the Medpor porous polyethylene implant. Aesthet Plast Surg 17:339–344View ArticleGoogle Scholar
- Frodel JL, Lee S (1998) The use of high-density polyethylene implants in facial deformities. Arch Otolaryngol Head Neck Surg 124:1219–1223View ArticlePubMedGoogle Scholar
- Haq J, Patel N, Weimer K, Matthews NS (2014) Single stage treatment of ankylosis of the temporomandibular joint using patient-specific total joint replacement and virtual surgical planning. Br J Oral Maxillofac Surg 52:350–355View ArticlePubMedGoogle Scholar
- Kozakiewicz M, Szymor P (2013) Comparison of pre-bent titanium mesh versus polyethylene implants in patient specific orbital reconstructions. Head Face Med 9:32View ArticlePubMedPubMed CentralGoogle Scholar
- Alonso-Rodriguez E, Cebrian JL, Nieto MJ, Del Castillo JL, Hernandez-Godoy J, Burgueno M (2015) Polyetheretherketone custom-made implants for craniofacial defects: report of 14 cases and review of the literature. J Craniomaxillofac Surg 43:1232–1238View ArticlePubMedGoogle Scholar
- O'Reilly EB, Barnett S, Madden C, Welch B, Mickey B, Rozen S (2015) Computed-tomography modeled polyether ether ketone (PEEK) implants in revision cranioplasty. J Plast Reconstr Aesthet Surg 68:329–338View ArticlePubMedGoogle Scholar
- Tesavibul P, Chantaweroad S, Laohaprapanon A et al (2015) Biocompatibility of hydroxyapatite scaffolds processed by lithography-based additive manufacturing. Biomed Mater Eng 26:31–38PubMedGoogle Scholar
- Garagiola U, Grigolato R, Soldo R et al (2016) Computer-aided design/computer-aided manufacturing of hydroxyapatite scaffolds for bone reconstruction in jawbone atrophy: a systematic review and case report. Maxillofac Plast Reconstr Surg 38:2View ArticlePubMedPubMed CentralGoogle Scholar
- Zhang L, Shen S, Yu H, Shen SG, Wang X (2015) Computer-aided design and computer-aided manufacturing hydroxyapatite/epoxide acrylate maleic compound construction for craniomaxillofacial bone defects. J Craniofac Surg 26:1477–1481View ArticlePubMedGoogle Scholar
- Figliuzzi M, Mangano FG, Fortunato L et al (2013) Vertical ridge augmentation of the atrophic posterior mandible with custom-made, computer-aided design/computer-aided manufacturing porous hydroxyapatite scaffolds. J Craniofac Surg 24:856–859View ArticlePubMedGoogle Scholar
- Essig H, Lindhorst D, Gander T et al (2017) Patient-specific biodegradable implant in pediatric craniofacial surgery. J Craniomaxillofac Surg 45:216–222View ArticlePubMedGoogle Scholar
- Xue AS, Koshy JC, Weathers WM et al (2014) Local foreign-body reaction to commercial biodegradable implants: an in vivo animal study. Craniomaxillofac Trauma Reconstr 7:27–34View ArticlePubMedPubMed CentralGoogle Scholar
- Wiltfang J, Merten HA, Becker HJ, Luhr HG (1999) The resorbable miniplate system Lactosorb in a growing cranio-osteoplasty animal model. J Craniomaxillofac Surg 27:207–210View ArticlePubMedGoogle Scholar
- Klammert U, Gbureck U, Vorndran E, Rodiger J, Meyer-Marcotty P, Kubler AC (2010) 3D powder printed calcium phosphate implants for reconstruction of cranial and maxillofacial defects. J Craniomaxillofac Surg 38:565–570View ArticlePubMedGoogle Scholar
- Zhao L, Patel PK, Cohen M (2012) Application of virtual surgical planning with computer assisted design and manufacturing technology to cranio-maxillofacial surgery. Arch Plast Surg 39:309–316View ArticlePubMedPubMed CentralGoogle Scholar
- Danelson KA, Gordon ES, David LR, Stitzel JD (2009) Using a three dimensional model of the pediatric skull for pre-operative planning in the treatment of craniosynostosis—biomed 2009. Biomed Sci Instrum 45:358–363PubMedGoogle Scholar
- Daniel M, Watson J, Hoskison E, Sama A (2011) Frontal sinus models and onlay templates in osteoplastic flap surgery. J Laryngol Otol 125:82–85View ArticlePubMedGoogle Scholar
- Hatamleh MM, Cartmill M, Watson J (2013) Management of extensive frontal cranioplasty defects. J Craniofac Surg 24:2018–2022View ArticlePubMedGoogle Scholar
- Azuma M, Yanagawa T, Ishibashi-Kanno N et al (2014) Mandibular reconstruction using plates prebent to fit rapid prototyping 3-dimensional printing models ameliorates contour deformity. Head Face Med 10:45View ArticlePubMedPubMed CentralGoogle Scholar
- Lim CG, Campbell DI, Clucas DM (2014) Rapid prototyping technology in orbital floor reconstruction: application in three patients. Craniomaxillofac Trauma Reconstr 7:143–146View ArticlePubMedPubMed CentralGoogle Scholar
- Liu YF, Xu LW, Zhu HY, Liu SS (2014) Technical procedures for template-guided surgery for mandibular reconstruction based on digital design and manufacturing. Biomed Eng Online 13:63View ArticlePubMedPubMed CentralGoogle Scholar
- Soleman J, Thieringer F, Beinemann J, Kunz C, Guzman R (2015) Computer-assisted virtual planning and surgical template fabrication for frontoorbital advancement. Neurosurg Focus 38:E5View ArticlePubMedGoogle Scholar
- Park SW, Choi JW, Koh KS, Oh TS (2015) Mirror-imaged rapid prototype skull model and pre-molded synthetic scaffold to achieve optimal orbital cavity reconstruction. J Oral Maxillofac Surg 73:1540–1553View ArticlePubMedGoogle Scholar
- Man QW, Jia J, Liu K, Chen G, Liu B (2015) Secondary reconstruction for mandibular osteoradionecrosis defect with fibula osteomyocutaneous flap flowthrough from radial forearm flap using stereolithographic 3-dimensional printing modeling technology. J Craniofac Surg 26:e190–e193View ArticlePubMedGoogle Scholar
- Yamada H, Nakaoka K, Horiuchi T et al (2014) Mandibular reconstruction using custom-made titanium mesh tray and particulate cancellous bone and marrow harvested from bilateral posterior ilia. J Plast Surg Hand Surg 48:183–190View ArticlePubMedGoogle Scholar
- Kim BJ, Hong KS, Park KJ, Park DH, Chung YG, Kang SH (2012) Customized cranioplasty implants using three-dimensional printers and polymethyl-methacrylate casting. J Korean Neurosurg Soc 52:541–546View ArticlePubMedPubMed CentralGoogle Scholar
- Li J, Hsu Y, Luo E, Khadka A, Hu J (2011) Computer-aided design and manufacturing and rapid prototyped nanoscale hydroxyapatite/polyamide (n-HA/PA) construction for condylar defect caused by mandibular angle ostectomy. Aesthet Plast Surg 35:636–640View ArticleGoogle Scholar
- Rotaru H, Schumacher R, Kim SG, Dinu C (2015) Selective laser melted titanium implants: a new technique for the reconstruction of extensive zygomatic complex defects. Maxillofac Plast Reconstr Surg 37:1View ArticlePubMedPubMed CentralGoogle Scholar
- Jo YY, Kim SG, Kim MK, Shin SH, Ahn J, Seok H (2017) Mandibular reconstruction using a customized three-dimensional titanium implant applied on the lingual surface of the mandible. J Craniofac Surg. https://doi.org/10.1097/SCS.0000000000004119